Type: KS-Arrow™ UHFAuD
Ultra High Frequency – Au Coating (Detector Side)
|Resonance Frequency||2000 kHz||700 - 2000 kHz|
|Mean Width||42 µm|
|Thickness||0.7 µm||0.5 - 0.9 µm|
Premounted AFM probes for use with Keysight Technologies Quick Scan Nose Cones
NanoWorld® Arrow™ ultra-high frequency AFM probes are capable of resonating with a very high frequency of up to 2.0 MHz. This AFM probe type combines outstanding sensitivity with fast scanning ability. All AFM probes of the Arrow™ series are made from monolithic silicon which is highly doped to dissipate static charge. They are chemically inert and offer a high mechanical Q-factor for high sensitivity. The AFM probes feature an AFM cantilever with a triangular free end and a tetrahedral AFM tip with a height of 3 µm.Additionally, this AFM probe offers an AFM tip radius of curvature of less than 10 nm.
The unique Arrow™ shape with the AFM tip position at the very end of the AFM cantilever allows easy positioning of the AFM tip on the area of interest.If needed, specific AFM cantilever thicknesses can be selected within very narrow tolerances for an additional fee.
A trapezoidal cross section of the AFM cantilever and therefore 30% wider (e.g. NCH) AFM cantilever detector side result in easier and faster laser adjustment. Additionally, because there is simply more space to place and reflect the laser beam, a higher SUM signal is reached.
Tip shape: Arrow
Gold Reflex Coating
The gold reflex coating consists of a 35 nm thick gold layer deposited on the detector side of the AFM cantilever which enhances the reflectance of the laser beam by a factor of 2.5. Furthermore it prevents light from interfering within the AFM cantilever.
As the coating is almost stress-free the bending of the AFM cantilever due to stress is less than 2 degrees.
For more information contact: firstname.lastname@example.org
Pointprobe® is a registered trademark of NanoWorld AG
All data are subject to change without notice.
Rue des Saars 10
For detailed information about our AFM probe product series please see below: