How to perform an AFM measurement in tapping, acoustic or non-contact mode on Bruker® AFM systems with Nanoscope® software, using CDT-NCLR AFM tips - NanoWorld®
This is a how-to-use step-by-step instruction on how to set up and scan in tapping, acoustic or non-contact mode using the respective NanoWorld® AFM probe.
This instruction is valid for the following NanoWorld® tapping mode AFM probes:
- Arrow™ NC
- Arrow™ NCR
How to optimize your scan parameters? Click here for more information!
Bruker® and Nanoscope® are trademarks of Bruker Corporation
For detailed information about our AFM probe product series please see below: